PQ001-Fiber NMR Analyzer was launched in 2008. After years of upgrade, PQ001 has many advantages such as small size, high precision, good repeatability, good stability and excellent cost/benefit characteristics. PQ001 has been widely used in the determination of oil content of textile fiber.
- Magnet: permanent magnet 0.5±0.08T
- Probe: Ø25mm
- Effective sample detection area:Ø25mm×H35mm;
- Size (L, W, H): 1685mm×520mm×386mm
- Weight: 134Kg
- Rapid determination of Spin Finish content of textile fiber;
- Oil range: 0.1%-100%
- Test accuracy: 0.1% error for 5% oil content; 0.02% error for 0.5% oil content;
- Repeatability: Relative standard deviation less than 2% (0.5% < oil content < 1%);
- Test limitation: 0.01% for 1 gram of sample;
- Compliance with national standards: GB/T 6504-2008;
- RSTD<=4.000% (tested in 565 days)
- User friendly Windows XP operation system and easy-to-use software; operating personnel can operate easily even with no training; multi-language.